The following information is for the EMANT300, EMANT380
Measure Temperature using Thermistor
Thermistors are widely used because of their sensitivity, small size, ruggedness and low cost. Thermistors have an electrical resistance that varies nonlinearly with temperature. The RT characteristics of most thermistors can be described by the SteinhartHart equation:
1/T = A + B*(Ln R) + C*(Ln R)^{3}
T is the absolute temperature (in Kelvin) and A, B, and C are constants which can be determined by measuring three sets of resistance and temperature values during calibration.
Most thermistors have a negative temperature coefficient (NTC), their resistance decreases with increasing temperature. Thermistors are specified according to its nominal resistance at 25 ^{o}C and commonly available thermistors range from 250 ohms to 100 kohms
The thermistor that we are using has the following characteristics

Nominal resistance @ 25^{ o}C: 10 kohms

negative temperature coefficient (NTC)

SteinhartHart equation parameters:

A= 0.001129148

B= 0.000234125

C= 8.76741E8

As the DAQ module Analog Input measures only voltage, we will need to provide a current source to convert the resistance to voltage. The EMANT300 has an 8 bit current DAC (digital to analog converter). As the DAC has 8 bits resolution, we can drive the resistance from 0 to 1mA in 255 steps with increments of about 39uA.
To measure temperature using the thermistor with the EMANT300 DAQ Training Kit,

Connect the thermistor to the Light Application Adaptor screw terminals labeled IDAC and AGND

Connect a wire from IDAC to AIN3

Connect a wire from AGND to AIN2
The LabVIEW VI that reads the voltage across the thermistor, converts the voltage to resistance and then temperature is called EMANT300 Example Thermistor.VI
Front Panel
Diagram
In the formula node, the first line calculates the Thermistor resistance. The next line converts from resistance to temperature in Kelvin and finally converts the temperature from Kelvin to Centigrade.